- Open Access
Validation of a flow cytometry based chemokine internalization assay for use in evaluating the pharmacodynamic response to a receptor antagonist
© Wyant et al; licensee BioMed Central Ltd. 2008
- Received: 03 September 2008
- Accepted: 01 December 2008
- Published: 01 December 2008
Pharmacodynamic assays are important in clinical trial design to investigate the relationship between drug concentration (pharmacokinetics) and drug "effect' or biological activity. Increasingly flow cytometry is being used to examine the pharmacodynamic effect of new drug entities. However, to date, the analytical validation of cytometry based assays is limited and there is no suitable guidance for method validation of flow cytometry-based pharmacodynamic assays. Here we report the validation of a flow cytometry-based chemokine internalization assay for use in evaluating the effect of a receptor antagonist in clinical trials. The assay method was validated by examining the stability of the reagent, assay robustness, sensitivity, repeatability and reproducibility precision. Experimental results show the assay reagent was stable over 26 weeks. The assay demonstrated a sensitivity to distinguish 0.005 μg/ml of a CCR2 antagonist with a %CV of 13.3%. The intra-assay repeatability was less than 15% with an inter-assay repeatability of less than 20%. In vivo study results demonstrated that the assay was consistent and a reliable measure of antagonist activity.
- Pharmacodynamic Effect
- Internalization Assay
- Flow Cytometry Assay
- CCR2 Antagonist
- Freeze Thaw Cycle
Chemokines are a class of small proteins that have potent chemotactic activity for cells of the immune system. In addition, they have the ability to activate leukocytes, to stimulate cytokine and proteolytic enzyme production, to mediate angiogenesis, and may be involved in cell proliferation and death.  The chemokine receptor CCR2 is widely expressed on mononuclear cells and a subset of memory (CD45RO+) CD4+ helper T cells. Activation of CCR2 by monocyte chemoattractant protein-1 (MCP-1), the major CCR2 ligand, is known to mediate chemotaxis and degranulation of monocytes as well as migration of activated effector memory T cells. [2, 3] The MCP-1/CCR2 pathway has been implicated in a variety of disease states such as Rheumatoid Arthritis, Multiple Sclerosis, and Atherosclerosis making the development of antagonists of this pathway an attractive pharmacological target [4–8]. Currently several companies have begun clinical trials of CCR2 antagonists .
In vitro pharmacodynamic assays are increasingly being utilized to demonstrate that a compound is having a desired biological effect after in vivo dosing. For CCR2 antagonists, the monitored effect is inhibition of either receptor signaling or ligand binding, depending on the mode of action of the drug being examined. When bound to their receptors chemokines, such as MCP-1, induce the receptor to internalize [2, 3]. We have capitalized on this and developed a flow cytometry assay to measure MCP-1 ligand internalization in clinical trials as a measure of the pharmacodynamic effect of our CCR2 antagonist.
Unlike pharmacokinetic and immunogenicity assays [10–15], there has not been any regulatory guidance published on the essential parameters for validation of pharmacodynamic assays such as those based on flow cytometry. In the past, variations in instruments, instrument settings, reagents and population heterogeneity had made validating assays based on flow cytometry difficult. Fortunately, advances in instrument standardization protocols based on fluorescent beads, more user friendly instruments and a greater reagent and instrument control by manufacturers has now made it possible to address the criteria and rigor that would accompany a validated flow cytometry assay . Using the guidance for ligand binding assays  as a foundation in which to base the validation of a flow cytometry pharmacodynamic assay and applying the "appropriate" parameters for a cell based cytometry assay, we validated a MCP-1 internalization assay. The parameters we examined included the stability of the reagents, the robustness, sensitivity, repeatability, precision and reproducibility of the assay. The precision was determined both in the in vitro validation phase and through retrospective analysis of in-study data.
Generation of Alexa 488 labeled MCP-1
Recombinant carrier free human MCP-1 was purchased from R&D Systems and fluorescently labeled with Alexa 488 (In Vitrogen Molecular Probes) using the conditions recommended for small proteins by Molecular Probes procedure. Alexa Fluor 488 was chosen due to the dye's increased stability and resistance to pH changes over a wide range of pH values (InVitrogen). The Alexa-488 labeled MCP-1 (AF488-MCP-1) was purified from the excess labeling reagent and free MCP-1 by RP-HPLC using a Vydac C18 semi-prep column (10 × 250 mm) and BioCad Vision Workstation. Labeled Peaks were identified and examined for their ability to bind to receptor positive cells. Identified peaks were pooled and retested for binding in a flow cytometry binding assay. The reagent was aliquoted, tested for freeze-thaw stability and frozen at -70°C.
MCP-1 alexa 488 internalization assay
For the purpose of assay validation, whole blood collected from normal healthy volunteers was incubated ex-vivo either with or without the CCR2 antagonist prior to the addition of the fluorescent staining reagents. MESF (Mean Equivalence of Soluble Fluorescence) values were determined by utilizing standardized MESF calibration beads (Bangs Laboratories Fishers, IN).
Overall criteria for evaluation
In general, for determination of %CV relevance, the guidelines established for ligand binding ELISA pharmacokinetic assays  was used to establish the %CV boundaries. A %CV less than 20% was considered an acceptable parameter. A 25% CV was used for values falling on the lower ends of curves. It was expected that fractional values such as that observed after saturation inhibition to have greater variability. Similarly, the inter-person variability was also anticipated to be greater and was to be documented here by retrospective analysis of phase one data.
Instrument set-up, MESF calibration and data analysis
A Becton Dickenson FACSCalibur instrument using 488 argon and red-diode lasers was calibrated daily using QC3 calibration beads (Bangs Laboratories). MESF was determined using the Quantum 1000 series bead sets from Bangs Laboratories daily. All raw instrument data was analyzed using WinList 5.0 (Verty Software House). Curve fitting and determination of EC50 and IC50 values was performed using Prism 4.0 (Graphpad) when applicable. The mean, standard deviation, standard error and % coefficient of variation (%CV) were calculated using Excel 2003 (Microsoft).
Reagent titration on whole blood
In order to determine the optimum reagent AF488-MCP-1 concentrations to use in the assay, a titration curve was performed. Serial dilutions of AF488-MCP-1 was added to 80 μL of whole blood and allowed to incubate at 37°C for 1 hour. The maximum internalization at 1 hour was determined to be the point at which no additional fluorescence was observed with increasing concentrations of AF488-MCP-1. For purposes of the in vitro validation, titration curves were generated by serially diluting a CCR2 antagonist into whole blood and incubating at room temperature for one hour prior to addition of the AF488-MCP-1. The CCR2 antagonist used here was an in house anti-CCR2 antibody which had been demonstrated to inhibit the binding and activity of MCP-1 in vitro (data not shown).
AF488-MCP-1 reagent stability was determined by examining both the binding of AF488-MCP-1 in whole blood over time and after five freeze-thaw cycles of the reagents. Stability of the AF488-MCP-1 was measured over a 26 week period at -70°C. Stock reagent stored at -70°C was diluted down to 150 nM, 100 nM, and 50 nM and added to whole blood (final concentration of AF488-MCP: 15 nM, 10 nM, 5 nM). Four different healthy volunteer blood donors were tested in the internalization assay at each time point and the resulting MESF and % positive values from each individual were averaged. Freeze-thaw (-70°C) stability was assessed by aliquoting the AF488-MCP-1 and cycling the various aliquots through different numbers of freeze-thaws. The cycled AF488-MCP-1 was subsequently utilized in the internalization assay and the resultant values for each cycle compared. Since the material was frozen after production the 1st freeze thaw cycle is the baseline value from which all other freeze thaw values were compared.
Robustness and sensitivity
Assay robustness was defined as how "reproducibly" the assay performed over time within the same blood sample, or in other words, how well the assay can withstand deliberate manipulation of environmental influences. Since the whole blood samples were to be shipped to a processing site, robustness was addressed by assaying the internalization of bound AF488-MCP-1 in CD14 (+) and CD4+CDR45RO+ cells over time at 1 hour, 24 hours, 48 hours and 72 hours after in vitro spiking of whole blood samples. Changes in overall fluorescence or the percentage of cells able to internalize MCP-1 were compared to the one hour "fresh" sample. Additionally, in order to examine the effect of overnight shipping on inhibition of internalization of AF488-MCP-1 by the receptor antagonist a direct comparison of the effect of overnight storage on the IC50 of the CCR2 inhibitor was examined. Briefly, receptor antagonist was incubated with whole blood at ambient temperature for 24 hours followed by processing through the internalization assay. Results obtained from the overnight incubation were compared to results obtained by processing the whole blood after only one hour incubation with the CCR2 antagonist.
The sensitivity of the assay, or the ability of the assay to demonstrate inhibition of ligand internalization at low concentrations of CCR2 inhibitor, was determined by serially diluting the CCR2 antagonist into whole blood and incubating for 1 hour at room temperature. The curves generated from the results of a minimum of 4 individuals were averaged.
Assay reproducibility was determined by assaying internalization of AF488-MCP-1 from the 10 different donors' blood drawn at 3 different times (each individual drawn 3 times). The blood draws were spaced 2–4 days apart to allow for recovery of the donor prior to the next blood draw. Measuring the intra-individual donor repeatability was accomplished by performing the internalization assay in triplicate. The mean, standard deviation and % CV were calculated from triplicate values (intra-sample repeatability), for each individual over time (intra-person reproducibility/inter-assay repeatability), and across individuals (inter-person reproducibility).
108 individuals (54 placebos, 54 CCR2-antagonist treated) were assayed in the internalization assay over a 113 day in the absence (placebos) or presence (treated) of AF488-MCP-1 the CCR2 receptor antagonist. Volunteers were dosed with a single dose of either antagonist or vehicle control and whole blood was drawn, shipped overnight to the processing laboratory and assayed. Blood samples were drawn prior to dosing (pre) and immediately (within 5 minutes) following completion of the infusion on Day 1, and again on day 3 (9 individuals only), 8, 15, 29, 43, 57, 71, 85, and 113. All 108 (54 placebo and 54 dosed) individuals were assessed at all time points except day 3. Mean, standard deviation % CV and standard error for the data grouped across all placebos and placebos + pre-dose of all 108 individuals were examined. The pharmacodynamic effect was examined by plotting the internalization of AF488-MCP-1 in CD14+ monocytes and memory helper T cells (CD4+CD45RO+) after dosing with the CCR2 antagonist on the first day. The pharmacodynamic effect in the dosed group was measured throughout the period however; the pharmacokinetic/pharmacodynamic relationship is beyond the scope of this manuscript.
Freeze thaw stability of AF488-MCP1
Assay robustness and sensitivity
It was determined that due to extremely low fluorescence values using MESF as an analytical measure on memory T cells (CD4+CD45RO+) cells was not reproducible (MESF range on CD4+CD45RO+ cells was 0 to 617 MESF with an average of 320 as compared to CD14+ MESF range of 53054 to 75851 averaging 61853 MESF). For this reason MESF values for the memory T cell population are not reported.
Percent Inhibition of monocyte internalization of AL488-MCP-1 with a CCR2 inhibitor
CCR2 inhibitior (μg/ml)
Reproducibility of Monocyte MESF
Reproducibility of Percentage of Memory T cells Positive
Average % CV of triplicate test of 10 individuals over 3 days
In study intra and inter-person variability
All Pre-dose samples
Percentage of Memory T cell
All Placebo over 113 days
The inter-person reproducibility of the assay was demonstrated by testing the pre-dose samples of all 108 individuals. The average fluorescence MESF value for monocytes was 101586.5 ± 38366.8 across all pre-dose samples (Table 6). The observed variation in the fluorescence values for the CD14+ cells across the population was 37.8%. The percentage of CD4+CD45RO+ T cell staining positive for AF488 MCP-1 internalization was 22.7 ± 0.7 giving a 3.2%CV for all of the individual pre-dose values.
The ability to demonstrate biological activity of drug (pharmacodynamic biomarkers) has become a valuable measurement in the development cycle of a pharmaceutical. For example, changes in lipid profiles, CRP and blood pressure have been used as pharmacodynamic (PD) measures for the assessment of drug treatment effect [17–21]. Pharmacodynamic assays such as the one described here are important for the overall clinical development of a pharmaceutical entity for which target effects are not easily identified in vivo. The ability to confidently and reliably demonstrate the action of drug on target enables the researcher early on to evaluate drug effects. Unfortunately, many of the mechanistic pharmacological effects have been difficult to observe in vivo and/or have been limited by the availability of reliable techniques for measuring the effect.
Flow cytometry has been a proven useful tool in the diagnosis of hematological disorders  as well as the diagnosis and monitoring of progression for diseases such as AIDS  and hematological cancers . Recent advances in instrument platforms, reagent quality and the increase in clinical usage have driven flow cytometry to be highly reproducible and consistent. For these reasons, flow cytometry has become an excellent platform for pharmacodynamic assays.
Cytometry can measure both phenotypic and functional parameters from whole blood. To date, the use of flow cytometry in clinical trials has for the most part been limited to measuring changes in phenotypic profiles and cell populations in response to therapeutics. However, several examples of flow cytometry assays being used to measure PD effects have recently been published. Ebo et al have recently published the validation of a flow cytometry assay showing the antagonistic effect of a compound on the ability of basophils to undergo a shape change . Similarly, Kelly et al has demonstrated the effect of an anti-CD40 antibody on the levels of circulating B cells in cynomolgus monkey . While the examples noted have demonstrated the power and versatility of flow cytometry in clinical and pre-clinical trials, to date, there has been no guidance or white papers released addressing validation of pharmacodynamic assays, particularly assays based on flow cytometry. This is in contrast to other assays used in drug development such as ligand binding and immunogenicity assays [10–15]. Additionally, there has been a recent white paper published in which recommendations were made on the use and validation of conventional biomarker assays in clinical trials [27, 28]. To date, no parameters for validating specifically flow cytometry PD assay have been proposed.
These guidance and white papers can serve as a template or guide for the validation of a flow cytometry PD assay. Flow cytometry PD assays should also demonstrate similar parameters: reagent stability (to ensure reagents remain consistent from day to day), assay robustness (how the assay is affected by variables such as overnight shipping), sensitivity (how the assay responds to low concentrations of compound i.e. the drug effect threshold), repeatability (the variability of results when the assay is performed multiple times on the same sample (inter-assay) and in replicate (intra-assay)) and reproducibility precision (the variability of results from blood drawn from the same individual multiple times and also from different individuals). Accuracy, as classically defined for ELISA and mass spectrometry based technologies, is the ability of an assay to measure analyte against a standard or quantitative control (QC) and is the most difficult of the parameters to define for flow cytometry based assay because the controls that would allow one to address this parameter directly (such as fixed cells of known antigen expression and density) do not exist or are untried for this purpose. However, several bead based methods by which instrument precision can be measured do exist. These bead based procedures not only standardize instrument settings accounting for daily fluctuations but also include fluorescence intensity standards to which to relate the results allowing greater reproducibility of data from instruments across laboratories and over time . The lack of standard controls for each parameter examined in flow cytometry limits the ability of the assay to be used in a quantitative manner and at best renders it semi-quantitative. For this reason, analysis in longitudinal clinical trials may best be served by examining the value observed relative post exposure to a baseline pre-drug treatment value. It is therefore critical that the variability of the cytometric assay be well understood prior to initiation of a clinical trial. Further refinement of the longitudinal variability during phase 1 trials solidifies our confidence in the assay and allows the scientist to better define the limits of what observed effects can be considered as a direct result of the therapy and not due to variations in the assay.
The purpose of validating an assay is to be able to demonstrate that the assay not only scientifically addresses the questions (Fit-for-Purpose ) but also does so in a reproducible and reliable manner. To this end, a whole blood alexa-488 labeled MCP-1 internalization assay was validated here for use in clinical trials investigating a CCR2 antagonist by examining a standard reagent concentration to use, the stability of the reagent, the robustness of the assay, the reproducibility across individuals over time, and the intra-assay repeatability through replicate analysis. Since this is a whole blood assay other parameters such as matrix effects were not examined However, parameters such as these are important to consider when moving from differing disease states and may require cross validations in new disease state whole blood matrix [12, 27].
The AF488-MCP-1 internalization assay described here was shown to be sensitive, robust, repeatable and reproducible. The assay is able to demonstrate a pharmacodynamic effect after in vivo dosing and additionally established that expression of CCR2 on both monocytes and memory T cells is relatively stable over 113 days using a controlled flow cytometry platform. The AF488-MCP-1 internalization assay will be an important tool which allows the clinical researcher to determine CCR2 saturation and a pharmacodynamic-pharmacokinetic relationship in clinical trials investigating CCR2 ligand antagonists.
The authors would like to thank Brain Harty for is invaluable work on this project.
- Szekanecz Z, Koch AE: Chemokines and angiogenesis. Curr Opin Rheumatol. 2001, 13 (3): 202-208.View ArticlePubMedGoogle Scholar
- Needham M, Sturgess N, Cerillo G, Green I, Warburton H, Wilson R, Martin L, Barratt D, Anderson M, Reilly C, Hollis M: Monocyte chemoattractant protein-1: receptor interactions and calcium signaling mechanisms. J Leukoc Biol. 1996, 60 (6): 793-803.PubMedGoogle Scholar
- Dzenko KA, Andjelkovic AV, Kuziel WA, Pachter JS: The chemokine receptor CCR2 mediates the binding and internalization of monocyte chemoattractant protein-1 along brain microvessels. J Neurosci. 2001, 21 (23): 9214-23.PubMedGoogle Scholar
- Quinones MP, Estrada CA, Kalkonde Y, Ahuja SK, Kuziel WA, Mack M, Ahuja SS: The complex role of the chemokine receptor CCR2 in collagen-induced arthritis: implications for therapeutic targeting of CCR2 in rheumatoid arthritis. J Mol Med. 2005, 83 (9): 672-81.View ArticlePubMedGoogle Scholar
- Dawson J, Miltz W, Mir AK, Wiessner C: Targeting monocyte chemoattractant protein-1 signalling in disease. Expert Opin Ther Targets. 2003, 7 (1): 35-48.View ArticlePubMedGoogle Scholar
- Barlic J, Murphy PM: Chemokine regulation of atherosclerosis. J Leukoc Biol. 2007, 82 (2): 226-36.View ArticlePubMedGoogle Scholar
- Schober A, Zernecke A: Chemokines in vascular remodeling. Thromb Haemost. 2007, 97 (5): 730-7.PubMedGoogle Scholar
- Szczucinski A, Losy J: Chemokines and chemokine receptors in multiple sclerosis. Potential targets for new therapies. Acta Neurol Scand. 2007, 115 (3): 137-46.View ArticlePubMedGoogle Scholar
- Xia M, Hou C, Demong DE, Pollack SR, Pan M, Brackley JA, Jain N, Gerchak C, Singer M, Malaviya R, Matheis M, Olini G, Cavender D, Wachter M: Synthesis, Structure-Activity Relationship and in Vivo Antiinflammatory Efficacy of Substituted Dipiperidines as CCR2 Antagonists. J Med Chem. 2007, 50 (23): 5561-5563.View ArticlePubMedGoogle Scholar
- DeSilva B, Smith W, Weiner R, Kelley M, Smolec J, Lee B, Khan M, Tacey R, Hill H, Celniker A: Recommendations for the bioanalytical method validation of ligand-binding assays to support pharmacokinetic assessments of macromolecules. Pharm Res. 2003, 20 (11): 1885-900.View ArticlePubMedGoogle Scholar
- Findlay JW, Smith WC, Lee JW, Nordblom GD, Das I, DeSilva BS, Khan MN, Bowsher RR: Validation of immunoassays for bioanalysis: a pharmaceutical industry perspective. J Pharm Biomed Anal. 2000, 21 (6): 1249-73.View ArticlePubMedGoogle Scholar
- Smolec J, DeSilva B, Smith W, Weiner R, Kelly M, Lee B, Khan M, Tacey R, Hill H, Celniker A, Shah V, Bowsher R, Mire-Sluis A, Findlay JW, Saltarelli M, Quarmby V, Lansky D, Dillard R, Ullmann M, Keller S, Karnes HT: Bioanalytical method validation for macromolecules in support of pharmacokinetic studies. Pharm Res. 2005, 22 (9): 1425-31.View ArticlePubMedGoogle Scholar
- Viswanathan CT, Bansal S, Booth B, DeStefano AJ, Rose MJ, Sailstad J, Shah VP, Skelly JP, Swann PG, Weiner R: Quantitative bioanalytical methods validation and implementation: best practices for chromatographic and ligand binding assays. Pharm Res. 2007, 24 (10): 1962-73.View ArticlePubMedGoogle Scholar
- Bansal S, DeStefano A: Key elements of bioanalytical method validation for small molecules. AAPS J. 2007, 9 (1): E109-14.PubMed CentralView ArticlePubMedGoogle Scholar
- Gupta S, Indelicato SR, Jethwa V, Kawabata T, Kelley M, Mire-Sluis AR, Richards SM, Rup B, Shores E, Swanson SJ, Wakshull E: Recommendations for the design, optimization, and qualification of cell-based assays used for the detection of neutralizing antibody responses elicited to biological therapeutics. J Immunol Methods. 2007, 321 (1–2): 1-18.View ArticlePubMedGoogle Scholar
- Purvis N, Stelzer G: Multi-platform, multi-site instrumentation and reagent standardization. Cytometry. 1998, 33 (2): 156-65.View ArticlePubMedGoogle Scholar
- Davidson MH: Introduction: utilization of surrogate markers of atherosclerosis for the clinical development of pharmaceutical agents. Am J Cardiol. 2001, 87 (4A): 1A-7A.View ArticlePubMedGoogle Scholar
- Revkin JH, Shear CL, Pouleur HG, Ryder SW, Orloff DG: Biomarkers in the prevention and treatment of atherosclerosis: need, validation, and future. Pharmacol Rev. 2007, 59 (1): 40-53.View ArticlePubMedGoogle Scholar
- Devaraj S, Rogers J, Jialal I: Statins and biomarkers of inflammation. Curr Atheroscler Rep. 2007, 9 (1): 33-41.View ArticlePubMedGoogle Scholar
- Abramson JL, Vaccarino V: Pulse pressure and inflammatory process in atherosclerosis. Adv Cardiol. 2007, 44: 223-33.View ArticlePubMedGoogle Scholar
- Tsimikas S: Oxidized low-density lipoprotein biomarkers in atherosclerosis. Curr Atheroscler Rep. 2006, 8 (1): 55-61.View ArticlePubMedGoogle Scholar
- Drouet M, Lees O: Clinical applications of flow cytometry in hematology and immunology. Biol Cell. 1993, 78 (1–2): 73-8.View ArticlePubMedGoogle Scholar
- Pattanapanyasat K, Thakar MR: CD4+ T cell count as a tool to monitor HIV progression & anti-retroviral therapy. Indian J Med Res. 2005, 121 (4): 539-49.PubMedGoogle Scholar
- Di Noto R, Mirabelli P, Del Vecchio L: Flow cytometry analysis of acute promyelocytic leukemia: the power of 'surface hematology'. Leukemia. 2007, 21 (1): 4-8.View ArticlePubMedGoogle Scholar
- Ebo DG, Lechkar B, Schuerwegh AJ, Bridts CH, De Clerck LS, Stevens WJ: Validation of a two-color flow cytometric assay detecting in vitro basophil activation for the diagnosis of IgE-mediated natural rubber latex allergy. Allergy. 2002, 57 (8): 706-12.View ArticlePubMedGoogle Scholar
- Kelley SK, Gelzleichter T, Xie D, Lee WP, Darbonne WC, Qureshi F, Kissler K, Oflazoglu E, Grewal IS: Preclinical pharmacokinetics, pharmacodynamics, and activity of a humanized anti-CD40 antibody (SGN-40) in rodents and non-human primates. Br J Pharmacol. 2006, 148 (8): 1116-23.PubMed CentralView ArticlePubMedGoogle Scholar
- Lee JW, Devanarayan V, Barrett YC, Weiner R, Allinson J, Fountain S, Keller S, Weinryb I, Green M, Duan L, Rogers JA, Millham R, O'Brien PJ, Sailstad J, Khan M, Ray C, Wagner JA: Fit-for-purpose method development and validation for successful biomarker measurement. Pharm Res. 2006, 23 (2): 312-28.View ArticlePubMedGoogle Scholar
- Colburn WA, Lee JW: Biomarkers, validation and pharmacokinetic-pharmacodynamic modelling. Clin Pharmacokinet. 2003, 42 (12): 997-1022.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.