A novel human peristalsis-inspired 3D-printed gastroduodenal simulator to evaluate intragastric/duodenal metabolic devices: a proof-of-concept study
Journal of Translational Medicine volume 20, Article number: 149 (2022)
To the Editor,
Endoscopic bariatric and metabolic therapies (EBMT) is an attractive alternative to medical treatment and bariatric surgery owing to the efficacy and minimal invasiveness. Delay in gastric emptying or gastric retention with filling of the ingested food in stomach triggers satiation, and rapid gastric emptying is related to obesity [1, 2]. Therefore, delayed gastric emptying should be evaluated with an intragastric/duodenal device for EBMT to assess its performance in inducing weight loss in patients with obesity. However, no quantitative comparative results on the performance of delayed gastric emptying in intragastric balloons (IGBs), and gastro-duodenal flow restrictors (G-DFR) have been reported in a pre-clinical study. The aim of the present study was to introduce the 3D-printed gastroduodenal simulator that mimics human peristalsis as a virtual testing system for the quantitative measurement of delayed gastric emptying in novel intragastric/duodenal EBMT devices before animal trials can be conducted. We also planned a pilot animal study whether this novel device with substantial delayed gastric emptying in experimental porcine group may affect relative weight loss compared with control porcine group.
The geometry and dimensions of the gastroduodenal simulator were designed to match the shape of the human stomach, based on computed tomography gastrography images of a healthy volunteer from a previous study . The simulator was fabricated using 3D-printed molds (Omg SLA; Xiamen, China), elastomer (Ecoflex 00-30; Smooth-on, Inc., USA), and silicone adhesive (Sil-Poxy; Smooth-on, Inc.). The simulator comprised four segments (fundus, body, antrum, and pylorus-duodenal bulb), each of which was paired with two actuators (L16-R Miniature Linear Servos; Actuonix, British Columbia, Canada) to contact the stomach body. To simulate peristalsis, eight motors moving in a straight line were mounted to generate motility in these regions. The contraction wave (α), which controls the stroke parameter of peristalsis, was defined as follows: α = diameter in contraction/diameter in relaxation. Each region was arranged with a relative contraction wave (α) of 0.9, 0.6, 0.6, and 0.2, respectively. The frequency parameter of peristalsis was set to twice per minute. Measurements in the simulator with peristalsis were obtained at different stages of drained and residual gastric material to assess gastric emptying volume defined as the partly drained fluid mass collected and retention ratio measured residual markers in the stomach. The normal range of values for gastric retention and gastric emptying after 2 h following a liquid meal were obtained from a prior study of 10 asymptomatic healthy volunteers in which it was defined as 25.9% ± (2 × 12.5%) . This was consistent with the retention ratio of 28.61% we observed in our control experiment using the simulator with peristalsis for 4 h. (Additional file 1: Video S1).
The gastric emptying volume of experimental model occupying G-DFR with 60-mm polytetrafluoroethylene (PTFE) skirt was the least gastric emptying volume suggesting enhanced gastric retention (percentage of emptying volume: 32.4%, retention: 49.2%) compared with IGB (percentage of emptying volume: 47.9%, retention: 37.9%), and G-DFR with a 30-mm PTFE skirt (percentage of emptying volume: 41.7%, retention: 35.0%). (Fig. 1) To clarify the quantitative measurement of delayed gastric emptying with favorable results on G-DFR with 60 mm distal PTFE skirt in this simulator with peristalsis, exploratory porcine study was performed. (Fig. 2A) This pilot porcine study was only evaluated the short-term outcomes (28-day and 42-day follow up) in terms of body weight and G-DFR migration. In the exploratory porcine (Yorkshire pig, 35–40 kg, 2 in experimental and 2 in control group) study, experimental porcine group with a novel G-DFR showed relative weight loss relative to the control group but without statistical significance: − 0.7% vs. 1.4% in 3 days; 5.2% vs. 15.7% in 14 days; 16.3% vs. 32.9% in 28 days; and 25.3% vs. 48.0% in 42 days. (Fig. 2B) No proximal or distal migration of a novel G-DFR was observed in the experimental group. (Additional file 2: Video S2).
In summary, we could quantitatively define the delayed gastric emptying volume for intragastric and gastroduodenal metabolic devices and evaluate the differences between these devices, which led to a delayed gastric emptying. The promising relative weight loss of our preliminary porcine study may suggest that this 3D-printed peristaltic simulator platform may predict the performance of a novel G-DFR with optimal delayed gastric emptying for EBMT.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
Endoscopic bariatric and metabolic therapies
Gastro-duodenal flow restrictors
Wright RA, Krinsky S, Fleeman C, Trujillo J, Teague E. Gastric emptying and obesity. Gastroenterology. 1983;84(4):747–51.
Abu Dayyeh BK, Acosta A, Camilleri M, et al. Endoscopic sleeve gastroplasty alters gastric physiology and induces loss of body weight in obese individuals. Clin Gastroenterol Hepatol. 2017;15(1):37-43.e31.
Kwon J, Choi J, Lee S, et al. Modelling and manufacturing of 3D-printed, patient-specific, and anthropomorphic gastric phantoms: a pilot study. Sci Rep. 2020;10(1):18976.
Maetani I, Ukita T, Tada T, et al. Gastric emptying in patients with palliative stenting for malignant gastric outlet obstruction. Hepatogastroenterology. 2008;55(81):298–302.
This work was supported by a Korea Medical Device Development Fund grant of the Korean government (the Ministry of Science and ICT, the Ministry of Trade, Industry and Energy, the Ministry of Health & Welfare, the Ministry of Food and Drug Safety) (Project Number: KMDF_PR_20200901_0266, 1711138598), by Basic Science Research Program through the National Research Foundation of Korea (NRF) funded by the Ministry of Education (2021R1A6A1A03040260) and by a grant (2021IP0015) from the Asan Institute for Life Sciences, Asan Medical Center, Seoul, Korea.
Ethics approval and consent to participate
This study was approved by the Institutional Animal Care and Use Committee (IACUC no. 2021-12-049) as it accorded with its guidelines the humane handling of laboratory animals.
Consent for publication
Do Hyun Park is a listed inventor on an issued patent for a gastric–duodenal flow restrictor (G-DFR) owned by the University of Ulsan Foundation for Industry Cooperation and the Asan Foundation. The authors have no financial relationships with any commercial entity producing healthcare-related products and/or services relevant to this article.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Kwon, J., Bang, C.S., Kim, S.O. et al. A novel human peristalsis-inspired 3D-printed gastroduodenal simulator to evaluate intragastric/duodenal metabolic devices: a proof-of-concept study. J Transl Med 20, 149 (2022). https://doi.org/10.1186/s12967-022-03357-z
- Gastric emptying
- Endoscopic bariatric and metabolic therapies
- 3D-printed gastroduodenal simulator