Detection of tumor-associated cells in cryopreserved peripheral blood mononuclear cell samples for retrospective analysis
© The Author(s) 2016
Received: 11 December 2015
Accepted: 21 June 2016
Published: 2 July 2016
Cryopreserved peripheral blood mononuclear cells (PBMCs) are commonly collected in biobanks. However, little data exist regarding the preservation of tumor-associated cells in cryopreserved collections. The objective of this study was to determine the feasibility of using the CellSieve™ microfiltration assay for the isolation of circulating tumor cells (CTCs) and circulating cancer-associated macrophage-like cells (CAMLs) from cryopreserved PBMC samples.
Blood samples spiked with breast (MCF-7), prostate (PC-3), and renal (786-O) cancer cell lines were used to establish analytical accuracy, efficiency, and reproducibility after cryopreservation. The spiked samples were processed through Ficoll separation, and cryopreservation was followed by thawing and microfiltration.
MCF-7 cells were successfully retrieved with recovery efficiencies of 90.5 % without cryopreservation and 87.8 and 89.0 %, respectively, on day 7 and day 66 following cryopreservation. The corresponding recovery efficiencies of PC-3 cells were 83.3 % without cryopreservation and 85.3 and 84.7 %, respectively, after cryopreservation. Recovery efficiencies of 786-O cells were 92.7 % without cryopreservation, and 82.7 and 81.3 %, respectively, after cryopreservation. The recovered cells retained the morphologic characteristics and immunohistochemical markers that had been observed before freezing. The protocols were further validated by quantitation of CAMLs in blood samples from two patients with renal cell carcinoma (RCC). The recovery rates of CTCs and CAMLs from cryopreserved samples were not statistically significant different (P > 0.05) from matched fresh samples.
To our knowledge, this is the first report that CAMLs could be cryopreserved and analyzed after thawing with microfiltration technology. The application of microfiltration technology to cryopreserved samples will enable much greater retrospective study of cancer patients in relation to long-term outcomes.
KeywordsCirculating cancer-associated macrophage-like cells (CAMLs) CellSieve™ microfiltration Cryopreservation Circulating tumor cells (CTCs) Ficoll Fluorescence antibody staining
Circulating tumor-associated cellular structures, such as circulating tumor cells (CTCs) and circulating cancer-associated macrophage-like cells (CAMLs) in blood, have emerged as important prognostic and predictive biomarkers for diagnosis, prognosis, and management of cancer diseases [1–6]. A blood based biopsy has advantages over traditional tissue biopsy, including lower cost, reduced risk, and repeatability. These assays are usually performed on fresh blood samples within a limited timeframe after blood draw. Little is known about the effects of cryopreservation on the recovery and characterization of all types of tumor-associated cells. Furthermore, assays compatible only with fresh samples require clinical on-site processing, because many factors, including sample transportation, age of blood, temperature fluctuations, and preservative reagents, can affect assay performance and reproducibility.
Prospective cryopreservation is a common method for long-term storage of biospecimens and is an invaluable tool for retrospective clinical studies of selected populations [7–9]. However, these biospecimens may be incompatible with assays that require intact cell membranes, because of ice crystal formation and subsequent lysis during the thawing process. Thus, assays compatible with current cryopreservation methods would allow the retrospective analysis of biomarkers in patient cohorts with known clinical outcomes. Additional advantages include (1) performing the assay in batches in a centralized laboratory to minimize interlaboratory variations and (2) assaying longitudinal collections together to minimize interassay variations.
Microfiltration is an effective method for isolating CTCs and other cells of interest from several types of cancers [1–3, 10–14]. CellSieve™ microfiltration technology, developed by Creatv MicroTech, Inc, uses a novel filter membrane with a high density of evenly distributed pores (160,000 pores per filter) with precisely controlled pore size (7 µm) to achieve high capture efficiency of tumor cells and low contamination of blood cells. The membrane material has low background fluorescence that allows the cells to be further characterized through various fluorescence-based assays.
The objective of our study was to determine whether CellSieve™ microfiltration technology can be applied to recovery and analysis of CTCs and CAMLs from cryopreserved PBMC samples. Initially, we used blood samples spiked with three cancer cell lines, MCF-7, PC-3, and 786-O, to demonstrate technical feasibility of the system. The protocols were validated by analyzing CAMLs recovered from cryopreserved PBMC samples of patients with renal cell carcinoma (RCC).
Cell lines and reagents
786-O (renal cell carcinoma), MCF7 (breast cancer), PC-3 (prostate cancer), SKBR3 (breast cancer), MDA-MB-231 (breast cancer), LNCaP (prostate cancer), and primary HUVEC (human umbilical vein endothelial cells) cell lines and others as previously described [1, 2, 15] were all obtained from American Type Culture Collection (ATCC). The cell lines were cultured using conditions specified by ATCC protocols. Cell suspensions were prepared with standard trypsin-treatment methods. Total cell counts were determined manually with a hemocytometer. Cell viability was assessed with trypan blue dye exclusion assay. Fresh cells with viability greater than 98 % were used for spiking experiments. CellSieve™ CTC Enumeration Kits (Creatv MicroTech, Inc) were used for recovery and fluorescence antibody staining of the filter-captured cells. These kits contain CellSieve™ filter membranes and reagents required for microfiltration of blood and/or PBMC samples and characterization of the filter-captured cells, including ready-to-use prefixation buffer, postfixation buffer, permeabilization buffer and mounting solution with DAPI (4′,6-diamidino-2-phenylindole), and antibody cocktail mixtures at optimized concentrations. To stain breast or prostate cancer cells, antibody mixture containing cytokeratin (CK) 8, 18, 19/fluorescein (FITC), epithelial cell adhesion molecule (EpCAM)/phycoerythrin (PE), CD45/Cyanine 5 was used. To stain RCC cells, antibody mixture containing CK8, 18, 19/FITC, vimentin/eFluor® 615 (EF615) and CD45/Cyanine 5 was used.
Patient eligibility and recruitment
Subjects eligible for enrollment in the Multidisciplinary Genitourinary Diseases Biospecimen Bank were those seen at Mayo Clinic’s campus in Scottsdale, Arizona, and were ≥18 years of age, able to provide informed consent, and undergoing evaluation to determine eligibility as a kidney donor or for treatment of genitourinary diseases. During the course of routine clinical visits at Mayo Clinic, subjects were consented to participate in the Biospecimen Bank. The protocol for collecting biospecimens was approved under a Mayo Clinic Institutional Review Board (IRB) approved protocol (08-000980). For healthy control subjects, blood samples were collected at Creatv MicroTech from healthy donors signed informed consent under a protocol approved by the Western IRB.
CellSieve™ microfiltration platform
Compared with conventional microfiltration methods, the CellSieve™ microfilter membrane provides (i) precise uniform pore size and distribution for maximal filtration efficiency, (ii) high porosity for fast filtration process, (iii) low background allowing high quality images, (iv) flat surface on the microscope slide minimizing the need for refocusing during microscopy, and (v) strong material to avoid breakage. To eliminate unnecessary handling procedures, such as filter disassembly for immunofluorescence, a filter cartridge is designed to aid in processing sample microfiltration followed by on-cartridge assay procedures; all of the assay procedures including antibody staining can be completed within the cartridge. With multiple cartridges, multiple blood samples can be analyzed simultaneously. Furthermore, the use of a programmable syringe pump provides consistent flow rate. The overall sample processing time is less than 2.5 h.
Microfiltration of healthy blood samples spiked with cell lines
Samples of whole blood were collected from healthy donors into sodium heparin Vacutainer tubes (Becton, Dickinson and Co). Separation of PBMC was performed with Ficoll-Paque (GE Healthcare) according to the protocols used in Mayo Clinic’s biobank for processing clinical samples . 7.5 mL of whole blood was centrifuged at 1600 rpm for 15 min. Plasma was removed and replaced with the same volume of phosphate buffered saline (PBS). A cell suspension was prepared by 1:1 diluted the cell sample with PBS. Ten mL of Ficoll-Paque solution (GE-Healthcare) was added in a 50 mL conical tube. The diluted cell suspension was then slowly and carefully layered on the top of Ficoll-Paque solution without mixing. The sample was centrifuged at 400×g for 30 min at 20 °C in a swinging bucket rotor with brake-off. Immediately after centrifugation, the upper layer was aspirated and discarded. The mononuclear cell layer was carefully transferred to a new 15-mL conical tube and mixed with PBS to a total of 15 mL. The sample was centrifuged at 300×g at room temperature for 10 min with brake-on. The supernatant was removed and discarded. The PBMC was gently washed one more time with PBS. The cells pellet was resuspended in 1 mL of cryogenic medium (10 % dimethyl sulfoxide and 90 % fetal bovine serum), and transferred into a Nunc (Sigma-Aldrich Co LLC) cryovial. The tubes were placed on Mr. Frosty tube racks and immediately placed in dry ice (less than 10 min at room temperature). The cryovials on the rack were directly stored at −80 °C freezer overnight. The frozen cryovials were transferred and placed in liquid nitrogen if it was needed.
For spiking experiments, a defined number of live tumor cells were spiked into the blood samples. The spiked samples were subjected to Ficoll separation to isolate fractions containing PBMCs and tumor cells. The mononuclear cells were washed, suspended in 1 mL of cryogenic medium, and transferred into a Nunc cryovials (1 mL/vial). For each cell line, four vials of sample were prepared. One vial was used as a control, with no cryopreservation. The other three vials were stored at −80 °C until thawed. After a period of storage, the frozen samples were quickly thawed in a 37 °C water bath, followed by immediate processing through CellSieve™ microfiltration in accordance with Creatv MicroTech’s protocol. Briefly, the filter membrane was rinsed with 5 mL of PBS. Each of the cell samples was washed, prefixed, and then filtered through the filter membrane (flow rate, 5 mL/min). The filter membrane was washed five times with PBS. The cells on the membrane were further treated with postfixation and permeabilization buffers (Creatv MicroTech, Inc.).
To stain recovered MCF-7 and PC-3 cells, we added 150 µL of fluorescent antibody mixture, against cytokeratins 8, 18, 19/FITC, EpCAM/PE, and CD45/Cyanine5, and incubated the samples at room temperature for 1 h. To stain recovered 786-O cells, fluorescent antibody mixture against cytokeratins 8, 18, 19/FITC, Vimentin/EF615, and CD45/Cyanine5 was used. Unbound antibodies were washed away with PBS.
After completion of the assay, the filter cartridge was disassembled and the filter membrane removed and placed onto a clean microscope slide, then mounted with 10 µL of mounting solution with DAPI and a cover slip for microscopy examination. The positivity of each marker was defined for any cells with a relative fluorescence signal greater than threefold over the background. The majority of cells were the single cells. For a cluster of small number of cells that were aggregated into a clump, we counted each cluster as one cell. The recovered tumor cells were counted from five fields of view under the 10× objective covering about 5 % of the filter area, and an average was calculated for estimating the total number of cells on the entire filter membrane.
Microfiltration of blood samples collected from patients affected by metastatic renal cell carcinoma
Whole blood samples, with three matched tubes for each blood draw, were collected in CellSave tubes (Janssen Diagnostics, LLC) from patients with RCC at Mayo Clinic and shipped to Creatv MicroTech for analysis. The concordance of the recovery was determined through processing of the matched tubes as with cryopreservation and without cryopreservation, respectively. Two of the tubes labeled as without cryopreservation (Tube 1 and Tube 2) were processed through CellSieve™ microfiltration within 24 h after blood draw. The third tube was processed through Ficoll separation, cryopreservation at −80 °C for 7 days, followed by thawing and CellSieve™ microfiltration, and antibody staining. The procedures of microfiltration and antibody staining were the same as used in the spiking experiments. The filter-captured cells from the Tubes 1 and 3 were stained with an antibody mixture consists of fluorescent dye-conjugated antibodies against CK8, 18, 19/FITC, vimentin/EF615 and CD45/Cyanine5. The filter-captured cells from the Tube 2 were stained with an antibody mixture consists of fluorescence-conjugated antibodies against PD-L1/Alexa Fluor 488, vimentin/EF615 and CD45/Cyanine5. The CTC was defined as a nucleated cell with positive staining for CKs and vimentin but negative staining for CD45 for Tubes 1 and 3. The CAML was defined according to criteria described previously .
Recovery of spiked cell lines in cryopreserved human PBMC samples
Recovery efficiency of spiked tumor cell lines in cryopreserved human PBMC samples
Mean (SD) %
Cells recovered, no.
Cells recovered, no.
Cells recovered, no.
We obtained mean (standard deviation, SD) spike-in recovery rates of 89.1 (1.0) % for MCF-7, 84.4 (1.4) % for PC-3, and 85.6 (6.2) % for 786-O, respectively. Between the fresh and stored frozen samples, we observed that the cryopreservation/thaw procedure had no significant effect on the recovery efficiencies of MCF-7, PC-3 or 786-O cells (P > 0.05 by 2-way analysis of variance). A 10 % decrease in recovery rates of 786-O was observed from the frozen samples (Day 7, 82.7 %; Day 66, 81.3 %) compared with the fresh sample (Day 0, 92.7 %). The cell line results suggest that the cryopreservation procedure preserves CTCs for microfiltration with minimal loss.
To determine the lowest number of tumor cells that can be detected by the microfiltration system, a series of spiking experiments were performed using six representative cell lines (786-O, MCF7, PC-3, SKBR3, MDA-MB-231, LNCaP). In our spiking experiments, the lowest absolute number of cells detected in our system was 3, 9, 5, 5, 4, and 4 cell using 786-O, MCF7, PC3, SKBR3, MDA-MB231, and LNCaP cell lines, respectively.
The linearity and limit of tumor cell detection in the spiking experiments are shown in Additional file 1: Figure S1. To also determine if cells lacking genome wide chromosomal rearrangements or aneuploidy can be detected by the microfilitration system, we used primary HUVECs for spiking experiments. Similar to MCF-7, PC-3 and 786-O, our observed capture efficiency for HUVECs was greater than 94 % (Additional file 2: Table S1). Microfiltration did not impact the expression of endothelial markers (Additional file 3: Figure S2). We also investigated the effect of storage temperatures at −80 °C and in liquid nitrogen on PBMC samples. No significant differences in cell morphology and antibody staining patterns were observed between cells cryopreserved (at −80 °C) and in liquid nitrogen for three weeks (Additional file 4: Figure S3).
Expression of epithelial and mesenchymal biomarkers in processed cryopreserved tumor cell lines
Recovery and enumeration of CAMLs in RCC patient samples processed with and without cryopreservation
Comparison of RCC patient blood samples processed with or without cryopreservation
Typical CAML, cells/tubea
Denucleated CAML, cells/tube
We did not detect statistically significant differences in CAML enumeration among the fresh vs frozen specimens (P > 0.05 by 2-way analysis of variance). We also measured the signal intensities of CKs in the cells from the Tube 1 and Tube 3. The Tube 2 was not used for analysis of signal intensity because it was not stained with CKs. All of the images for CKs were taken under a consistent exposure time (600 ms). Three cytoplasmic positions were selected for measurement of signal intensities of CKs in each CAML, representing the high, majority, and low signals of CKs. The CK signals were plotted versus each CAML cells (Additional file 5: Figure S4). The high and low signals of CKs were indicated as error bars. The background signal of the filter membrane, shown as a dotted line in the plot, was an average signal measured from the fourteen locations on the filter membrane. At Time Point 1, the majority of CK signal intensities in the fresh CAMLs (in range of 460–850, average 616.43 ± 97.72) were similar to that in the frozen CAMLs (range of 522–886, average 619.00 ± 100.71), suggesting that the CKs were preserved after the cryopreservation. At Time Points 2 and 3, the CK signals in the fresh CAMLs were slightly higher than that in the frozen CAMLs. We did not measure the signals of vimentin because different exposure times were used in the imaging to avoid signal saturation.
Denucleated CAMLs observed in both fresh and frozen RCC samples
Recovery of CTCs in RCC samples processed with and without cryopreservation
Analysis of tumor-associated cells in peripheral blood offers the advantage of a noninvasive approach to study changes in biomarkers associated with chemotherapy, tumor recurrence, or tumor treatment. However, many CTC technologies require the processing of fresh blood samples and are incompatible with cryopreserved samples, limiting the ability to perform retrospective studies on archived samples. This limitation may pose a challenge in multicenter clinical trials and require a central laboratory to process samples on the same day for the CTC assays. The assay reproducibility may be adversely affected by performing each fresh blood sample individually or by transportation of specimens between locations. Cryopreservation is one of the most commonly used methods for storage biologic specimens collected over time.
We optimized a cryopreservation work flow compatible with detecting tumor-associated cells in stored peripheral blood to overcome assay limitations that require processing of fresh blood. We showed the feasibility of CellSieve™ technology for isolation and identification of CTCs and CAMLs from cryopreserved samples. Our notable observations include (1) recovery of spiked tumor cells lines in cryopreserved peripheral blood using the CellSieve™ platform; (2) preservation of cellular morphologic characteristics and markers in cryopreserved samples from patients with metastatic RCC; and (3) preservation of CTC and CAML enumerations in samples cryopreserved up to 66 days at −80 °C.
When evaluating our observations, we found limitations to the present investigation. First, we analyzed 1 CTC microfiltration platform, and the cryopreservation procedure may not be compatible with other antigen-dependent or antigen-independent cell capture methods. Second, although we evaluated breast, prostate, and kidney cancer cell lines, we analyzed CTCs and CAMLs only in samples from patients with kidney cancer. The DMSO concentration or Ficoll separation may not be compatible with all types of cancer diseases, and further study is required to determine whether the cryopreservation procedure is suitable for other cancers. Third, we did not extend the study beyond 60 days, and the impact of storage beyond this time point may or may not have effect on enumeration assays.
Cryopreservation of PBMCs allows phenotypical and genetic assays to be performed in batches at a central laboratory and minimize assay variation. We have demonstrated that the CellSieve™ system is compatible with Ficoll separation and cryopreservation for detection of CTCs and CAMLs. Investigators have previously reported that CTCs can be recovered from cryopreserved PBMC samples [7, 9, 16]; however, it was unclear whether CAMLs can be recovered from cryopreserved PBMC sample. To our knowledge, we are the first to report that CAMLs can be separated by Ficoll method, cryopreserved for future use, and retrieved after thawing samples stored up to 60 days at −80 °C. In our Ficoll separation, CAMLs segregate to the interphase between Ficoll and plasma, where PBMCs and CTCs are located. CAMLs may have similar density to other mononuclear cells, such as lymphocytes and CTCs, allowing them to be enriched through Ficoll separation, and use of a microfiltration system can isolate cells on the basis of pore size. These enriched cells are suspended in cryoprotectant medium, which contains DMSO to reduce cell lysis associated with ice crystal formation. Preliminary analysis on the signal intensities of CKs between fresh and frozen CAMLs are similar, however, the CAML sample size is not large enough to draw definitive conclusions.
We observed from the matched samples that the freeze/thaw process has no adverse effect on morphologic characteristics and the biomarkers of RCC-associated cells. Although we did not observe statistically significant changes in enumeration of spiked breast and prostate cell lines, further study is required to evaluate whether our cryopreservation process is compatible with other tumor types. Furthermore, during the course of investigation, we also found denucleated CAMLs in frozen samples from a patient with RCC, consistent to fresh samples. The denucleated CAMLs resemble typical CAMLs closely in morphologic characteristics and staining patterns except for a lack of nuclei. However, the importance of these structures is undefined.
Tumor-associated cells can be recovered successfully from cryopreserved PBMC samples with CellSieve™ microfiltration. The morphologic features, biomarkers, and enumeration of tumor-associated cells are preserved in cryopreserved specimens. Application of CellSieve™ microfiltration for detection of tumor-associated cells in cryopreserved PBMC samples allows the biobanking of samples to be processed in batches, to minimize interassay variations, and also enables large-scale, long-term retrospective and epidemiologic analyses in selected cancer populations.
American Type Culture Collection
cancer-associated macrophage-like cells
circulating tumor cells
epithelial cell adhesion molecule
Institutional Review Board
peripheral blood mononuclear cells
phosphate buffered saline
programmed death-ligand 1
renal cell carcinoma
scanning electron microscope
PZ, MLS, EPC, RWJ, DLA, SL, PA, CMT, THH edited and wrote the manuscript. PZ, SL and DLA performed the immunofluorescence and microfiltration experiments. PZ, MLS, EPC, DLA and CMT designed the study. All authors read and approved the final manuscript.
The authors acknowledge the support provided by the Gloria A. and Thomas J. Dutson Jr., Kidney Research Endowment. This work was supported by the Mayo Clinic Center for Individualized Medicine (T.H.H.); a Gerstner Family Career Development Award (T.H.H., R.W.J.); Kathryn H. and Roger Penske Career Development Award to Support Medical Research (T.H.H.); the National Cancer Institute at the National Institutes of Health (NIH) (K12CA90628) (T.H.H.). The US Army Research Office (ARO) and the Defense Advanced Research Projects Agency (DARPA) (W911NF-14-C-0098) (C.-M.T.). This publication was made possible by Clinical and Translational Science Awards Grant Number UL1 TR000135 from the National Center for Advancing Translational Sciences (NCATS), a component of the NIH. Its contents are solely the responsibility of the authors and do not necessarily represent the official views of the NIH or the policy of the US Government.
PZ, DLA, SL, PA and CMT are employees of Creatv Microtech. MLS, EPC, RWJ and THH have no competing interests to declare. Mayo Clinic does not endorse the products mentioned in this article.
Availability of data and material
Data is available in the additional files and supplementary information. Cell lines are publicly available through American Type Culture Collection.
Consent for publication
Consent for publication was obtained through the Mayo Clinic IRB protocol (08-000980) with an institutional consent form.
Ethics approval and consent to participate
Subjects were consented to participate in the Biospecimen Bank. The protocol for collecting biospecimens was approved under a Mayo Clinic Institutional Review Board (IRB) approved protocol (08-000980). For healthy control subjects, blood samples were collected at Creatv MicroTech from healthy donors signed informed consent under a protocol approved by the Western IRB. No animal studies were performed.
The authors acknowledge the support provided by the Gloria A. and Thomas J. Dutson Jr, Kidney Research Endowment. This work was supported by the Mayo Clinic Center for Individualized Medicine (T.H.H.); a Gerstner Family Career Development Award (T.H.H., R.W.J.); Kathryn H. and Roger Penske Career Development Award to Support Medical Research (T.H.H.); the National Cancer Institute at the National Institutes of Health (NIH) (K12CA90628) (T.H.H.). The US Army Research Office (ARO) and the Defense Advanced Research Projects Agency (DARPA) (W911NF-14-C-0098) (C.-M.T.). This publication was made possible by Clinical and Translational Science Awards Grant Number UL1 TR000135 from the National Center for Advancing Translational Sciences (NCATS), a component of the NIH. The funders had no role in the design of the study and collection, analysis, interpretation of data or in writing the manuscript.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Adams DL, Zhu P, Makarova OV, Martin SS, Charpentier M, Chumsri S, Li S, Amstutz P, Tang CM. The systematic study of circulating tumor cell isolation using lithographic microfilters. RSC Adv. 2014;9:4334–42.View ArticlePubMedPubMed CentralGoogle Scholar
- Adams DL, Martin SS, Alpaugh RK, Charpentier M, Tsai S, Bergan RC, Ogden IM, Catalona W, Chumsri S, Tang CM, Cristofanilli M. Circulating giant macrophages as a potential biomarker of solid tumors. Proc Natl Acad Sci USA. 2014;111:3514–9.View ArticlePubMedPubMed CentralGoogle Scholar
- Adams DL, Adams DK, Stefansson S, Haudenschild C, Martin SS, Charpentier M, Chumsri S, Cristofanilli M, Tang CM, Alpaugh RK. Mitosis in circulating tumor cells stratifies highly aggressive breast carcinomas. Breast Cancer Res. 2016;18:44–9.View ArticlePubMedPubMed CentralGoogle Scholar
- Cohen SJ, Punt CJ, Iannotti N, Saidman BH, Sabbath KD, Gabrail NY, Picus J, Morse M, Mitchell E, Miller MC, Doyle GV, Tissing H, Terstappen LW, Meropol NJ. Relationship of circulating tumor cells to tumor response, progression-free survival, and overall survival in patients with metastatic colorectal cancer. J Clin Oncol. 2008;26:3213–21.View ArticlePubMedGoogle Scholar
- Cristofanilli M, Braun S. Circulating tumor cells revisited. JAMA. 2010;303:1092–3.View ArticlePubMedGoogle Scholar
- Cristofanilli M, Budd GT, Ellis MJ, Stopeck A, Matera J, Miller MC, Reuben JM, Doyle GV, Allard WJ, Terstappen LW, Hayes DF. Circulating tumor cells, disease progression, and survival in metastatic breast cancer. N Engl J Med. 2004;351:781–91.View ArticlePubMedGoogle Scholar
- Fluxion Biosciences, Inc. Cryopreservation of CTC samples for biobanking and sample storage. Revision 06-16-14. http://www.fluxionbio.com/isoflux-cryopreservation/. Accessed 2 May 2016.
- Ho TH, Nateras RN, Yan H, Park JG, Jensen S, Borges C, Lee JH, Champion MD, Tibes R, Bryce AH, Carballido EM, Todd MA, Joseph RW, Wong WW, Parker AS, Stanton ML, Castle EP. A multidisciplinary biospecimen bank of renal cell carcinomas compatible with discovery platforms at Mayo Clinic, Scottsdale, Arizona. PLoS One. 2015;10(7):e0132831. doi:10.1371/journal.pone.0132831.View ArticlePubMedPubMed CentralGoogle Scholar
- Lu D, Graf RP, Harvey M, Madan RA, Heery C, Marte J, Beasley S, Tsang KY, Krupa R, Louw J, Wahl J, Bales N, Landers M, Marrinucci D, Schlom J, Gulley JL, Dittamore R. Detection and characterization of circulating tumour cells from frozen peripheral blood mononuclear cells. J Circ Biomark. 2015;4:4. doi:10.5772/60745.Google Scholar
- De Giorgi V, Pinzani P, Salvianti F, Panelos J, Paglierani M, Janowska A, Grazzini M, Wechsler J, Orlando C, Santucci M, Lotti T, Pazzagli M, Massi D. Application of a filtration- and isolation-by-size technique for the detection of circulating tumor cells in cutaneous melanoma. J Invest Dermatol. 2010;130:2440–7.View ArticlePubMedGoogle Scholar
- Lin HK, Zheng S, Williams AJ, Balic M, Groshen S, Scher HI, Fleisher M, Stadler W, Datar RH, Tai YC, Cote RJ. Portable filter-based microdevice for detection and characterization of circulating tumor cells. Clin Cancer Res. 2010;16(20):5011–8.View ArticlePubMedPubMed CentralGoogle Scholar
- Vona G, Sabile A, Louha M, Sitruk V, Romana S, Schütze K, Capron F, Franco D, Pazzagli M, Vekemans M, Lacour B, Bréchot C, Paterlini-Bréchot P. Isolation by size of epithelial tumor cells: a new method for the immunomorphological and molecular characterization of circulating tumor cells. Am J Pathol. 2000;156:57–63.View ArticlePubMedPubMed CentralGoogle Scholar
- Xu T, Lu B, Tai Y-C, Goldkorn A. A cancer detection platform which measures telomerase activity from live circulating tumor cells captured on a microfilter. Cancer Res. 2010;70:6420–6.View ArticlePubMedPubMed CentralGoogle Scholar
- Zheng S, Lin H, Liu JQ, Balic M, Datar R, Cote RJ, Tai YC. Membrane microfilter device for selective capture, electrolysis and genomic analysis of human circulating tumor cells. J Chromatogr A. 2007;1162:154–61.View ArticlePubMedGoogle Scholar
- Adams DL, Stefansson S, Haudenschild C, Martin SS, Charpentier M, Chumsri S, Cristofanilli M, Tang CM, Alpaugh RK. Cytometric characterization of circulating tumor cells captured by microfiltration and their correlation to the Cellsearch(®) CTC test. Cytometry A. 2015;87:137–44.View ArticlePubMedGoogle Scholar
- Lu D, Harvey M, Madan R, Heery C, Marte J, Beasley S, Landers M, Krupa R, Louw J, Wahl J, Bales N, Marrinucci D, Schlom J, Gulley J, Dittamore R. Characterization of circulating tumor cells (CTCs) in cryopreserved metastatic castrate-resistant prostate cancer (mCRPC) patient samples. Proceedings: AACR 106th Annual Meeting 2015. 2015b; Philadelphia. (Abstract 1586).