Rat model of veno-arterial extracorporeal membrane oxygenation
© Ali et al.; licensee BioMed Central Ltd. 2014
Received: 29 October 2013
Accepted: 3 January 2014
Published: 7 February 2014
We aim to develop a rat model of veno-arterial extracorporeal membrane oxygenation (VA-ECMO).
VA-ECMO was established in twelve Male Sprague-Dawley rats (250-350 g) through cannulation of the right jugular vein for venous drainage and the right femoral artery for arterial reinfusion. Arterial blood pressure was measured using a conductance catheter through cannulation of the left carotid artery. Heart rate was monitored by electrocardiography and arterial blood gas parameters with a blood gas analyzer. The VA-ECMO circuit was tested by subjecting the rats to hypoxic cardiac arrest with resuscitation using VA-ECMO. Both load-dependent and load-independent measures of myocardial contractility were measured using pressure-volume loop analysis to confirm restoration of myocardial function post-resuscitation.
Following hypoxic cardiac arrest VA-ECMO provided sufficient oxygenation to support the circulation. The haemodynamic and blood gas parameters were maintained at transition and during ECMO. All animals were resuscitated, regained cardiac function and were able to be weaned off ECMO post-resuscitation.
We have established a safe, high-throughput, economical, functioning rat model of VA-ECMO.
KeywordsECMO Extracorporeal membrane oxygenation Resuscitation Cardiac arrest
Extracorporeal Membrane Oxygenation (ECMO) is a type of mechanical circulatory support that is used for resuscitation of patients with failing pulmonary and cardiac function . However, no effective rodent models of ECMO support following cardiac arrest currently exist to allow evaluation of post-arrest myocardial function. We describe a novel technique for instituting veno-arterial extracorporeal membrane oxygenation (VA-ECMO) in a rodent for resuscitation following hypoxic cardiac arrest. This method restores the native circulation allowing the opportunity to conduct further studies on the physiological and molecular changes occurring in the myocardium during and after cardiac arrest.
All animals received humane care in compliance with the 'Principles of laboratory animal care’ formulated by the National Society for Medical Research and the 'Guide for the care and use of laboratory animal resources’ published by the US National Institute of Health (NIH publication No. 85-23, revised 1996). The following studies were approved by the Columbia University Institutional Animal Care and Use Committee (IACUC) under protocol AC-AAAG6406 and conform to the Guide for the Care and Use of Laboratory Animals published by the National Institutes of Health.
Twelve 250-350 g male Sprague-Dawley rats were originally sourced from Charles River Laboratories for use in the experiments. Rats were maintained in temperature-controlled (20°C to 22°C) cages with a 12-hour light-dark cycle, and received free access to sterilized water and standard rodent chow (Rodent diet BK002P, B & K Ltd).
Anesthesia and ventilation
Rats were anesthetized using 3-4% isoflurane in an induction chamber and maintained at 1.5-2.5% through a nosecone. Endotracheal intubation was undertaken with a 14G cannula (Smiths Medical, London, UK).
The operative procedure, including implantation of all ECMO cannulae, was conducted in the live animal, prior to the induced hypoxic cardiac arrest. The animal was positioned with the lower limbs closest to the operating surgeon. Under visualization of a dissecting microscope an incision was made in the right groin to access the femoral vessels. The femoral artery was dissected free from the femoral vein, and the femoral vein was cannulated for administration of fluids and medication. One-hundred units of intravenous heparin were administered for anticoagulation.
A median sternotomy was performed to allow direct visualization of the heart for confirmation of cardiac arrest. The sternum was divided with electrocautery taking caution to avoid the right and left internal thoracic arteries. The chest was retracted open with a miniature Finochietto retractor and the thymus removed with electrocautery to facilitate visualization of the heart, lungs, and great vessels.
For ECMO outflow the right jugular vein was dissected in the cervical region and a small hemostatic clamp applied proximally to occlude the jugular vein. The vessel was then ligated and retracted as distal as possible above the clavicle to allow for cannulation. A small venotomy was made at the distal end of the jugular vein and a 20 G angiocath cannula connected to 4.0 mm tygon tubing advanced into the vein. A ligature was loosely placed around the jugular vein to secure the cannula and then the clamp occluding the vessel was removed. The cannula was then gently manipulated until it was positioned in the superior vena cava (SVC) under direct visualization.
Continuous acquisition of haemodynamic data was used to complement the procedure and allow for precise monitoring of arterial pressure before and after ECMO reperfusion. The right common carotid artery was dissected in the cervical region. The vessel was ligated as distally in the neck as possible, and a small occlusive clamp (Yasargil, Germany) was used to occlude the carotid artery proximal to the site of intended cannulation. A Millar ultra miniature pressure catheter was connected to a Millar-Ultra PVS console for measurement of left ventricular pressure over the course of the experiment. A small arteriotomy was made in the carotid artery distal to the clamp and the catheter was inserted. A 6.0 ligature was used to secure the catheter within the vessel.
Establishment of the extracorporeal membrane oxygenation circuit
The equipment and our ECMO model are depicted in Figure 1. A micro-peristaltic pump was utilized for propulsion within the circuit (Digital Reglo Pump, Ismatec, Ltd; Figure 1A, 1E). This is a digital tubing pump with 8 rotors. The pump is marketed for flows ranging from 0.002-57 ml/min with rotor speeds ranging from 1.6 to 160 revolutions per minute. The required prime volume for our ECMO model is 8 ml normal saline. 4.0 mm tygon tubing was inserted into a customized pump cassette (Ismatec, Ltd) and the outflow tubing arm was connected to a custom miniature oxygenator device (Figure 1C, 1F). Heparinized saline was used to prime the oxygenator and the circuit tubing. The pump inflow tubing was connected to the venous drainage cannula and the oxygenator was then connected to the arterial infusion cannula (Figure 1D). Finally a 12 ml syringe was added into the circuit proximal to the pump via a 3-way stopcock (Figure 1H) to act as a reservoir for addition of fluid if necessary to support pump flow, which can occur with changes in position of the venous cannula within the SVC. Arterial blood gases were measured using an Epoc blood analysis device (Epocal Inc, Ontario, Canada). The parameters of haemodynamics and blood gas analysis were recorded at 3 time points: preoperatively (0 mins), post cardiac arrest (15 mins), post resuscitation (60 min). Additional blood gas analysis were performed ad hoc post resuscitation to ensure correction of metabolic acidosis.
Data were subjected to the Kolmogorov-Smirnov test to determine distribution. Descriptive variables are presented as means ± SD and compared with the t-test. When comparing multiple groups data was analysed by analysis of variance with Bonferroni post-test for multiple comparison of parametric data. Estimations are presented with 95% confidence intervals. Conventional levels of significance (0.05) were applied throughout. Statistical analysis was undertaken using SPSS for windows version 19 (Chicago, Ill) and S Plus version 6 (Seattle, Wash).
Metabolic changes pre and post VA-ECMO resuscitation of hypoxic cardiac arrest
HCO3-, mmol/L (SD)
pCO2,, mmHg (SD)
pO2,, mmHg (SD)
Hb, g/dl (SD)
HCT, % (SD)
sO2, % (SD
K+, mmol/l (SD)
Ca+, mmol/l (SD)
Lactate, mmol/L (SD)
Assessment of haemodynamic and contractile function
Measures of myocardial contractility pre and post VA-ECMO resuscitation of hypoxic cardiac arrest
Load independent measurements
Ees (Emax), mmHg/μL (SD)
PRSW, mmHg (SD)
Load dependent measurements
dP/dt max, mmHg/s (SD)
dP/dt min, mmHg/s (SD)
Cardiac output, μl/min (SD)
Stroke volume, μl (SD)
Stroke work, mmHg/μl (SD)
Tau Glantz, msec (SD)
Well-functioning rat models of cardiopulmonary bypass have been established, including recent examples that do not require priming with blood . While these models are helpful, ECMO is fundamentally different than cardiopulmonary bypass in its clinical applications. The full bypass circuit is not applicable to the patient who needs long-term cardiopulmonary support. ECMO is applicable to a broad range of clinical scenarios, including bedside cannulation for support during instances of hypoxic cardiac arrest [3, 4]. A rabbit model of ECMO has recently been described to test these applications . Similar to the rabbit model, rats have a nearly identical anatomy to humans with the advantage of being smaller, less expensive, and easier to handle than larger animal models.
A small animal model for ECMO is necessary because it provides an efficient, economical, and accurate model for the study of physiologic changes during ECMO. The estimated cost of equipping a laboratory with a basic microsurgical setup such as dissecting microscope, small animal ventilator, and surgical instruments is approximately $3000. The costs specifically related to the ECMO circuit include micro-oxygenators ($250 each/4-6 uses), Tygon tubing ($200), intravenous cannuale ($200) and the micro-peristaltic pump ($2400). In our model, nearly all equipment is reusable and the cost incurred per experiment is largely related to the purchase and boarding costs of the rat (Male Sprague-Dawley rat, $30, Charles River Laboratories, Cambridge, MA; Boarding, approximately $20 for acclimatization and prompt usage).
In our study, we developed the first rodent model of VA-ECMO that was used for successful resuscitation following an induced hypoxic cardiac arrest. After a 30-minute period of supportive reperfusion, ECMO was weaned to assess the ability of the resuscitated heart to support the circulation independently. Based on our results, we were able to show that this model allows for return of cardiac function and serves as the basis for subsequent investigation in areas such as post-myocardial arrest depression, cardiogenic shock, hemodynamic support and cardiovascular resuscitation.
We have established a reliable and economical rat model for VA-ECMO. Biochemical parameters and pressure-volume relationships show that despite induced hypoxic cardiac arrest, our VA-ECMO circuit was capable of restoring adequate circulation for return of full cardiac function and perfusion.
(Extracorporeal membrane oxygenation)
(Veno-arterial extracorporeal membrane oxygenation)
(National Institutes of Health)
(Institutional Animal Care and Use Committee)
(Superior vena cava)
This project was supported by grants from the British Cardiovascular Society (AA), The Evelyn Foundation, Stanford University and Columbia University.
- Haneya A, Philipp A, Diez C: A 5-year experience with cardiopulmonary resuscitation using extracorporeal life support in non-postcardiotomy patients with cardiac arrest. Resuscitation. 2012, 83: 1331-1337. 10.1016/j.resuscitation.2012.07.009.View ArticlePubMedGoogle Scholar
- Zhu X, Ji B, Liu J: Establishment of a novel rat model without blood priming during normothermic cardiopulmonary bypass. Perfusion. 2013, 29: 63-69.View ArticlePubMedGoogle Scholar
- Mair P, Hoermann C, Moertl M, Bonatti J, Falbesoner C, Balogh D: Percutaneous venoarterial extracorporeal membrane oxygenation for emergency mechanical circulatory support. Resuscitation. 1996, 33: 29-34. 10.1016/S0300-9572(96)00989-6.View ArticlePubMedGoogle Scholar
- Robert Leeper W, Valdis M, Arntfield R, Ray Guo L: Extracorporeal membrane oxygenation in the acute treatment of cardiovascular collapse immediately post-partum. Interact Cardiovasc Thorac Surg. 2013, 17: 898-899. 10.1093/icvts/ivt358.PubMed CentralView ArticlePubMedGoogle Scholar
- Lu S, Pan S, Wang C, Hu K, Hong T: Establishment of an animal model of extracorporeal membrane oxygenation in rabbits. Perfusion. 2012, 27: 414-418. 10.1177/0267659112447653.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.